The DryView is an excellent seller worldwide thanks to its remark- able speed and outstanding image quality. Includes. • DryView (14”x17”) PACS . Kodak – Dryview Outstanding low- to midvolume performance in a value- priced solution, Kodak™ DryView™ Laser Imager provides outstanding low . The Kodak DryView laser imager is the right choice for health imaging facilities that need diagnostic-quality images but do not require high-volume output.
|Published (Last):||20 February 2009|
|PDF File Size:||5.2 Mb|
|ePub File Size:||13.63 Mb|
|Price:||Free* [*Free Regsitration Required]|
But they do contain silver and polyester that may be recovered by using one of several recycling processes. When you open the front door, the top sheets of film in the cartridge deyview be exposed. Page 22 Controls and Indicators Figure To print the manual completely, please, download it. A calibration film is printed whenever: Attention Message Area — Displays system status and error messages.
Page 64 Troubleshooting 3. If a film drvyiew at the cartridge prevents the cartridge from closing, error message P Press Open 5S will display. For a translation of the following warnings, please refer to the applicable safety addendum listed below: See error message on line 2 of the display.
Power Light — Indicates that the imager is powered up. Note If the front door is opened when the film cartridge is open, the top sheets of film in the cartridge are exposed. Press the Format Menu key to access the Format Menu. This equipment does not have a gas sealed electronics enclosure and could ignite any flammable or explosive gases present in its environment.
The numbers dryvview the illustration identify key areas on the path from film pickup to exposure through development and exit onto the film tray: Laser radiation may be present when the machine operates without the rear cover xryview.
In rare cases, film may stall in the transport area above the platen. Note When jams occur in Area 2, the film cartridge is left open. Section 5 includes system troubleshooting procedures. When an error message is displayed, perform the operator action listed in the table for that message. Troubleshooting Manually Closing the Film Cartridge The imager automatically closes the film cartridge when the Door Open key is pressed. Using the DryView V2 Keypad 6.
Caution Hot Surface Drum and rollers inside the processor are hot. There may be more than one sheet. See Removing Film Jams from Area 3 on page DryView Laser Imager User Guide Opening the Front Door Via drview Local Panel During normal machine operation, you will occasionally need to open the front door with power applied to load and unload film cartridges, or to change the charcoal filter.
Selecting the number of copies to print. Display window messages are explained in Section 6. The numbered descriptions below correspond to the circled numbers in Figure Electromagnetic compatibility EMC — If film is jammed in the drive rollers see Figurepull the green thumb tab to the left to open the rollers and free the film.
Dry view 8100 laser imager
And even with the most exacting procedures, the potential exists to discharge hazardous materials into the environment. P Contrast Req The user has requested a contrast test film. Film Throughput 55 films per hour assuming images of by pixels and a direct connect input.
Film Pickup Problems In Area 1 Troubleshooting Film Pickup Problems in Area 1 Area 1 identifies the location where film is removed from the film cartridge and positioned to be sent by drive rollers toward the exposure area see Figure Image Quality Barcode information on each film cartridge allows the DryView Laser Imager to automatically determine film type, the number of sheets remaining, and the film manufacturing lot.
Touch Screen — Displays a variety of menus used for system configuration and operation. P Press Open 5S The front door does not unlatch because the film When the odor become noticeable you should replace the filter. When you make a service call, be prepared to give the error message and message number to the call taker see page For keypad functions, refer to Section 4.
As the imager is used, the filter becomes less effective. Serial Number Location March Rev.
Kodak Dryview – Digital Radiographics
If the listed action does not clear the error message, call for service. Specifications Specifications Dimensions Height: For serial number location, see Figure Certificates dryvied Compliance and Declarations of Conformity have been issued as shown below. Display Brightness Set Keys — Increase or decrease the keypad display brightness.
If the imager is unable to close the cartridge, a P error displays.