The EC help provides an easy access to the information related to the use of Keysight EC Network Analyzer. Pressing Help key on the front panel. EB/EB, please see the Programming Manual. • Installation and Quick Start Guide (Part Number: Ex1, attached to optional ABA). This manual . The EC complies with INSTALLATION CATEGORY II as well as The Service Manual is a guide to servicing the EC ENA Series Network Analyzer .
|Published (Last):||17 April 2008|
|PDF File Size:||18.50 Mb|
|ePub File Size:||1.45 Mb|
|Price:||Free* [*Free Regsitration Required]|
You cannot use the front panel keys during the initial registration of the EC therefore connect the mouse and the keyboard before turning on the power. Figure Recover User Backup ecse Step 5. To use this function, you must prepare for recovery in advance. Therefore you should follow the procedure which you see on the EC. Italic type is used for emphasis. Do not replace components with the power cable connected.
In-position of push-button switch. Press any key to continue. To avoid injuries, always disconnect the power and discharge circuits before touching them.
You can also contact one of the following centers and ask for a test and measurement sales representative. Turn the screw in anti-clockwise direction until manul is locked. The touch screen calibration screen Figure appears. If the connector type of manua, standard of the calibration kit you use has polarity the distinction between male and femaleyou need to change the standard class definition of the calibration kit depending on the standard you actually use.
The printing date changes when a new edition is printed. Push the button under the lock and remove the hard disk. Turn the key to the left e501c it is locked.
Agilent Technologies EC Instruction manual |
Turn the key to the right until it is locked. Select the C drive to restore, then click Next button. EC firmware is executed, then the measurement display appears.
The Device Manager dialog box appears. This calibration makes it possible to perform measurements with the highest possible accuracy. A message box appears after e5071v completion of recovery process.
Atec Agilent-E5071C User Manual
The boot option is displayed. If the name label of the calibration kit has been changed, the label is displayed as the soft key. Select the destination folder and input a file name, then click Next button. If you do any operation other than the following steps, the system may not be recovered. Minor corrections and updates that are incorporated at reprint do not cause e5071d date to change.
Press the standby switch of the EC to turn it on. Press 1, and then press Enter. Files you created using the save function files in the D drive are not affected, but we recommend backing them up before executing system recovery for precautionary purposes.
Click Port x Open to start the calibration measurement x denotes the test port to which the standard is connected.
If the hard disk failed and has been replaced after purchase, the state when the replacement was performed is recovered. The manual part number changes when extensive technical changes are incorporated.
The error correction function will also be automatically enabled. Turn ON the EC. After e50771c seconds, the next screen appears automatically even if you do not press any key, so do not miss it.
In addition to the Windows operating system, the following settings of the EC are returned to the factory state. Remove the hard disk. Connect the keyboard to the EC. To use this function, you must create the user backup image in advance. F5071c message dialog box appears after mabual completion of the Recovery Image Setup procedure. Turn off the power. The HDD revision can be checked using the following procedure: Out-position of push-button switch.
Replacement Procedure Step 1.
E Removal Procedure Step 1. Agilent Technologies assumes no mnual for the customer’s failure to comply with these precautions. Click Short to start the calibration measurement. For more information, see Modifying Calibration Kit Definition.
CN or below Step 1. Follow the procedure described below to calibrate the touch screen.